Agilent technologies HP B Pdf User Manuals. View online or download Agilent technologies HP B Operating Manual. HP B Synthesizer Function Generator. Service Manual. Up for sale is the following Technical Manual in original binder/binding Original Manufacturer’s. I recently won a “for parts or repair” HP B function generator for a great Although free scans of the B service manual are readily.
|Published (Last):||18 October 2017|
|PDF File Size:||4.75 Mb|
|ePub File Size:||9.32 Mb|
|Price:||Free* [*Free Regsitration Required]|
HP B Synthesizer Function Generator Service Manual P/N | eBay
First verify that the output amplifier is causing the problem. Subject to credit approval. If not, go to Step g. Place the in Special Test Mode 54 by pressing the following keys: Modulation Source Block Diagram Explanation of the HP B frequency synthesis begins with Paragraph Phase Detector Observe the junction of R and R Severe over voltages may cause the fuse to open as well.
Select a valid country. If the frequency does not approach lOOkHz, troubleshoot the circuitry step c. Ramp reset may be initiated either by the sdrvice detector output Figure or by serrvice or — ramp reset signal from peak detectors at the output amplifier.
Please enter 5 or 9 numbers for the ZIP Code. Press Local several 4 or 5 times to quit.
Agilent Technologies HP 3325B Manuals
Output Amplifier Service Group K. Note and record the dc voltage reading on the volt- meter. Lamar, South Carolina, United States.
The triangle output frequency is the difference between the 1 MHz reference and the input frequency from the mixer divided by twenty. Using a peak detector, the 3325 sor is able to sample the D. Learn more – opens in new window or tab. This item will ship to United Statesbut the seller has not specified shipping options. If the test passes, plug in one connector at a time and re-run the test to determine the assembly causing this test to feul. Ramp polarity is determined by the ramp polarity gate.
Agilent Technologies 3325B Synthesizer/function Generator
Full text of ” HP: The maximum ratings are: You are covered by the eBay Money Back Guarantee if you receive an item that is not as described in the listing.
If amplitude problems are 3325b, it is important to disable this auto calibration.
Disconnect the signal and sync output cables from the front panel. This would require the VCO frequency to be Back to home page Return to top.
Special Functions 60 through 66 Will not sweep frequency. Basic Block Diagram of Control Circuits The charge slope is much greater than the discharge slope because the phase comparator current source has about ten times the magnitude of the 33325b current source. If the signal output is normal but there is no front panel sync output, check’for a square wave at both ends of the fuse, F4.